Monte Carlo N-Particle Transport Code Simulation
of Leksell Gamma Knife Using Disk 1 Sources of
Polystyrene, PMMA, Plastic Water and Head Phantom
Ma. Vanessa Francheska P. Perianes, Dorren Alexis F. Villanueva,
and Jade R. Dungao
Physics Department, De La Salle University
Taft Avenue, Manila, Philippines
Stereotactic Radiosurgery (SRS) has become a standard modality for the treatment of benign and metastatic brain lesions that were deemed medically unsuitable for surgery. The Leksell Gamma Knife (LGK), a type of SRS that was used in this study, has 201 Cobalt-60 sources distributed in a hemisphere whose radiation intersects at the isocenter. The relative dose at the isocenter was verified using Monte Carlo N-Particle Simulation (MCNP). This study uses disk sources, an alternative for the full geometry collimator system of the LGK, to simulate a 160-mm water phantom made of different materials: polystyrene, plastic water, and PMMA (Polymethyl methacrylate). In addition, the simulation of a head phantom was also included in this study. Relative dose distributions were calculated and were compared to the relative dose distributions from the cited literatures. As a result, no significant differences have been found. In conclusion, the use of disk sources provides a simpler method of simulating the LGK instead of using the full geometry collimator system in the MCNP Visual Editor.
Stereotactic Radiosurgery (SRS) is one of the treatment modalities used to treat benign and metastatic brain lesions. It is a specialized technique where a uniform dose, with highly conformal radiation, is delivered to the target.
The type of SRS that was used in this study was the Leksell Gamma Knife (LGK). LGK was developed by Swedish neurosurgeon Lars Leksell and Physicist Borje Larson. The LGK, manufactured by Elekta Instruments Inc., uses 201 Co-60 sources distributed on a hemisphere. The diameter of the beams of the ionizing radiation was defined by the collimator system and the shape of the target can be varied by changing the size of the collimator helmets: 4 mm, 8 mm, 14 mm and 18 mm. . . . . read more
CHEUNG JY, YU KN. 2006. Rotating and static sources for gamma knife radiosurgery systems: Monte Carlo studies. Med Phys. 33(7): 2500-2505.
MOSKVIN V, DESROSIERS C, PAPIEZ L, TIMMERMAN R,N RANDALL M, DESROSIERS P. 2002. Monte Carlo Simulation for the Leksell Gamma Knife: I. Source modelling and calculations in homogeneous media. Phys Med Biol 2002. 47(12):1995-2011
MCCONN JR, GESH CJ, PAGU RT, RUCKER RA, WILLIAMS III, RG. 2011. Compendium of Material Composition Data for Radiation Transport Modeling. Pacific Northwest National Laboratory, 2011. 1:18, 45, 54, 183, 242, 338.
SHULTIS, JK, FAW, RE. 2004. An MCNP Primer. Retrieved from http://www.mne.ksu.edu/`jks/MCNPprmr.pdf on 24 June 2016.
TRNKA J, KLUSON J. 2006. MCNP-based Computational Model for the Leksell Gamma Knife. Med Phys. 63-75p.
XIONG W, HUANG D, LEE L. 2007. Implementation of Monte Carlo Simulations for the Gamma Knife System. IOP Science.
YIPENG LI. 2002. Absorbed Dose Measurements and MCNP Modeling for the Leksell Gamma Knife. Retrieved from http://citeseerx.ist.psu.edu/viewdoc/download?doi=10.1.1.116.2752&rep=rep1&type=pdf on 24 June 2016
ZHU D, AUSTERLITZ C, BENHABIBS S, MOTA H, ALLISON RR, CAMPOS D. 2010. Study of a spherical phantom for Gamma Knife Dosimetry. Med Phys. Vol. 11.